iM3 Catalogue - Español - Flipbook - Página 57
Safe, Simple, Sterile
iM3 Straw - Maintenance Schedule
Leave full water bottle on the dental unit with water in
the lines at the end of the day. The Straw releases residual
disinfectant that is designed to be continually present in the
bottle and the lines.
Weekly (or to follow the weekend)
Remove bottle(s) and top off the water.
Ensure the Luer lock fitting is pushed completely into the
pick-up tube and threaded tight into the Straw. Improper
removal of the bottle from the unit can unseat the Luer lock.
We recommend the bottle be cleaned once a month.
If using an ultrasonic scaler, run the ultrasonic scaler for
sixty seconds. Due to the infrequent use of a scaler and the
minute amount of water used, the water inside the water
line may experience longer periods of stagnation.
Replace Straw with a new iM3 Straw and follow shock
treatment protocol written in the Straw directions.
Re-Order Online - Code PS001