iM3 Catalogue - Europe - Flipbook - Page 59
iM3 - The Veterinary Dental Company
Continuous delivery of clean, safe water for one year.
A safe to use, safe to handle, self-contained system.
Saves time and cost!
iM3 Straw - Maintenance Schedule
Leave full water bottle on the dental unit with water in the lines at the end of the day.
The Straw releases residual disinfectant that is designed to be continually present in the
bottle and the lines.
Weekly (or to follow the weekend)
Remove bottle(s) and top off the water.
Ensure the Luer Lock fitting is pushed completely into the pick-up tube and threaded
tightly into the Straw. Improper removal of the bottle from the unit can unseat the Luer
We recommend the bottle be cleaned once a month.
If using an ultrasonic scaler, run the ultrasonic scaler for sixty seconds. Due to the
infrequent use of a scaler and the minute amount of water used, the water inside the
water line may experience longer periods of stagnation.
Replace Straw with a new iM3 Straw and follow shock treatment protocol written in the
NEW iM3 Straw Ex
Now you can have all the great benefits of the iM3 Straw in a cartridge for use
with external water. Introducing the new iM3 Straw Ex and Fitting Kit (each sold
separately). Simply connect both ends of the fitting bracket to your water line
and attach the iM3 Straw Ex cartridge.
Works great on iM3 Pro S and other dental units that have external water lines.
Also works as a smart solution for non-iM3 dental units that don’t use standard
water bottles and therefore cannot work with the iM3 Straw ( PS001).
Note: While we expect you to get long lasting protection, the longevity of the
iM3 Straw Ex is affected by both the quality and volume of water that passes
For Veterinary Use Only
iM3 Straw Ex Cartridge
iM3 Straw Ex Fitting Kit