iM3 Catalogue - Europe - Flipbook - Page 75
iM3 | The Global Name in Veterinary Dentistry
Continuous delivery of clean, safe water for one year.
A safe to use, safe to handle, self-contained system.
Saves time and cost!
iM3 Straw - Maintenance Schedule
Leave full water bottle on the dental unit with water in the lines at the
end of the day. The Straw releases residual disinfectant that is designed
to be continually present in the bottle and the lines.
Weekly (or to follow the weekend)
Remove bottle(s) and top off the water.
Ensure the Luer Lock fitting is pushed completely into the pick-up tube
and threaded tightly into the Straw. Improper removal of the bottle
from the unit can unseat the Luer Lock.
We recommend the bottle be cleaned once a month.
If using an ultrasonic scaler, run the ultrasonic scaler for sixty seconds.
Due to the infrequent use of a scaler and the minute amount of water
used, the water inside the water line may experience longer periods of
Replace Straw with a new iM3 Straw and follow shock treatment
protocol written in the Straw directions.
Now you can have all the great benefits of the iM3 Straw in a cartridge for
use with external water. Introducing the new iM3 Straw Ex and Fitting Kit
(each sold separately). Simply connect both ends of the fitting bracket to
your water line and attach the iM3 Straw Ex cartridge.
Works great on iM3 Pro S and other dental units that have external water
lines. Also works as a smart solution for non-iM3 dental units that don’t
use standard water bottles and therefore cannot work with the iM3 Straw
Note: While we expect you to get long lasting protection, the longevity of
the iM3 Straw Ex is affected by both the quality and volume of water that
passes through it.
For Veterinary Use Only
iM3 Straw Ex Cartridge
iM3 Straw Ex Fitting Kit
NEW iM3 Straw Ex